Find great deals for National Instruments NI USB Low-cost Multifunction DAQ. Shop with confidence on eBay!. The NI USB is a USB based data acquisition (DAQ) and control device with analog input and output and digital input and output. The price is NOK. : National Instruments USB Bit, 48kS/s Low-Cost multifunction DAQ: Computers & Accessories.
|Published (Last):||27 April 2013|
|PDF File Size:||5.29 Mb|
|ePub File Size:||18.66 Mb|
|Price:||Free* [*Free Regsitration Required]|
Input high is between 2. We will perform a very simple test, which is sufficient if we are to check that both AO0 and AI0 work correctly. This document shows in detail how USB must be connected to external signals for measurement or control.
To prepare for the loopback test, AI0 and AO0 are wired together. Figure 7 shows MAX. Via the Polymorphic VI Selector you 60009 select the type of data the function will write.
Learn more – opens in new window or tab. The following relevant documents can be found at http: The configuration can be saved using the Save button, see Figure I got benefit from it.
By clicking the Finish button in the window shown in Figure 15, the window shown in Figure 16 is opened where you can select the physical channels for your task. Wire nuts may also be used. Output low is below 0.
Thanks for your prompt reply. Caq assumes all responsibility for this listing. Please enter 5 or 9 numbers for the ZIP Code. On-board voltage sources available at individual terminals: Depending on your system and noise levels, connecting pin 4 to ground may help. The latter is available on the CD shipped with the device, and can alternatively be downloaded for free from http: If you had written a value outside 66009 range to the DAQmx Write function you would get an error due to signal limitations function it can only generate voltage outputs between 0 and 5 volts.
I am sure that one of them is the signal but which one? Sell now – Have one to sell? Tested, fully functional, and in excellent physical condition. You are daw by the eBay Money Back Guarantee if you receive an item that is not as described in the listing.
The black wire is ground, What are the read and white wires? The pin connections you showed in your image are for differential inputs.
National Instruments USB Data Acquisition Card, NI DAQ, Multifunction | eBay
Mouse over to Zoom – Click to enlarge. To see the details about a virtual channel, you can click the Show Details buttoncf. Helpful to my needs. Also is this the correct wiring as shown in pics? When the wizard has finished the installation of the driver software, the USB is ready adq use after you click Finish Norwegian: Message 3 of This menu also has an import option.
USB – National Instruments
All these items, i. This seller offers shipping discounts on combined purchases for eligible items. See other items More Please enter a valid ZIP Code. Then we generate some legal voltage at AO0.
Get an immediate offer. The field attached to the bottom of the function is the Polymorphic VI Selector from which you can select the type of data the function will read. Message 6 of The wizard has finished the installation of the driver software, and the USB is ready for use, after you click Finish Norwegian: Connect 1 on the potentiometer to ground. P0 is port 0, and P1 is port 1.
Message 2 of Above we created a task for an analog input. I read your discussion. In the following only the name USB is used. Digital input DI and digital output DO: